I'd like to try my hand at building a few antennas for VHF/UHF use. I'm going to be going approximately from established designs, but obviously the final result will require some tuning. I don't currently own either an SWR meter or antenna analyzer. Is an SWR meter good enough for what I want to do, or will an antenna analyzer be necessary for this? What are the benefits of an antenna analyzer in such situations?
I've not personally built antennas from scratch, but I appreciate my antenna analyzer just for being a good instrument — making the invisible aspects of my antenna system visible.
Compared to using a SWR meter for the purpose, an antenna analyzer:
Displays more information.
A SWR meter still gives you enough information, in the sense that you can try things until they move towards 1:1 SWR, but it's a one-dimensional measurement: if you tried to adjust an antenna using it you're doing a “hotter colder” search, trying things until it improves.
An antenna analyzer will tell you resistance and reactance independently, which allows you to more directly understand the problem. For example, you can have an exactly resonant antenna, which simply has the wrong magnitude of impedance (you need an impedance transformer). On a SWR meter this can show up as the SWR having a minimum point but remaining well above 1:1. (That minimum point is not necessarily the resonant point.) On an antenna analyzer, you can immediately see that you've got R = 34 Ω instead of 50 Ω (or whatever), and build that correction into your balun.
High-end antenna analyzers can also give you measurements at multiple frequencies — either as a graph or individual readings at selected frequencies. A graph allows you to easily obtain measurements in the opposite direction — not “how suitable is this antenna at this frequency?”, but “what frequencies is this antenna good for?”, as well as other RF-electronics measurements like finding the resonant frequency of a circuit or the upper frequency limit of a ferrite choke.
In the above picture: my RigExpert AA-600 antenna analyzer displaying an impedance graph (it can also compute SWR from the same data). You can see directly from this graph that whatever it's connected to is resonant at the frequency at the middle of the graph (reactance crosses zero) and at that point has a resistance (and also magnitude of impedance) of 100 Ω.
Lets you use less transmitted power.
To measure the impedance of your antenna, no matter what instrument you use, you need a signal generator — a transmitter — somewhere. An antenna analyzer has a low-power signal generator built in.
If you use a SWR meter, you have to use a separate one; if you use your regular transceiver for the purpose, you're limited to its lowest power setting, which might be a few watts! Even if you have a lower power transmitter, it might not be sufficient to show a reading on the SWR meter, which has scales for much higher power levels.
By comparison, my antenna analyzer's output falls within FCC Part 15 emissions limits, so I'm much less likely to interfere with anyone else (and don't have to stop to identify myself, either).
An antenna analyzer is not necessary. Sure, it's handy. A VNA is even handier. For this equipment you're looking at something like $50 USD for a NanoVNA, up to many thousands of dollars for a lab grade VNA. If you just want to know if you need to trim a bit off a wire dipole or not, and simply get on the air without toasting your transmitter, none of these things are necessary. While there's a certain joy in being able to measure and quantify exactly how things are operating and make it all "just so", radios are in fact quite forgiving. An AC current wants to radiate. In fact, there's an entire industry around making things not radiate.
If you wonder how to make an antenna without these tools, you need only look to the recent past. It wasn't that long ago that we did not have solid state electronics, and test equipment like we have today was either very expensive, or nonexistent. Most of the methods used in those days to measure antenna impedance involve putting an antenna in some kind of bridge and comparing it against a known impedance.
Here's one example of a dead-simple approach:
You can read the full question for an explanation of how it works. It's not particularly sophisticated, but if you just need to trim your antenna until it's not a horrible match to 50 ohms, it will work just fine. If you have an SWR meter, you can also just adjust for minimum SWR and accomplish the same thing. Note you don't need the matching section (L1, L2, and C3) if you just want to measure impedance.
If you want to take the sophistication up a notch, you can use a grid dip meter. You can find them at any hamfest, or used on eBay or whatnot for some tens of dollars. Just a few decades ago, these were state of the art antenna test equipment. The physical laws that govern their operation have not changed, and they work as just well today.
As another step up, you can consider any number of used impedance bridges which are similar in concept to the schematic above, but have adjustable and calibrated inductors and capacitors so you can measure the complex impedance of the antenna. This is in effect what an antenna analyzer does, the only difference being that the antenna analyzer will automatically adjust the knobs for you, and maybe measure impedance at a range of frequencies at a push of button.
1$\begingroup$ your statement: "But then, these things could set you back thousands of dollars.". Sure if you buy lab equipment (maybe) but this is the hyperbolic statement I have ever read on any SE forum. I paid 124 (dollars) for the Autek VA1 and about 250 for the MFJ 259B. Maybe costly for some but NOT thousands of dollars! $\endgroup$– K7PEHJan 6, 2015 at 22:43
1$\begingroup$ @K7PEH good luck finding a working VNA for $124. $\endgroup$ Jan 6, 2015 at 23:48
$\begingroup$ @phil and now they start at $50 :) $\endgroup$– tomnexusAug 28, 2020 at 2:59
1$\begingroup$ True, nanovna has lowered the price of admission significantly. I've updated the answer. $\endgroup$ Aug 28, 2020 at 3:19
I couldn't live without my two antenna analyzers -- I may even get a VNA (vector network analyzer) although that is not necessary for simple antenna analysis.
I own both the MFJ 259B and the Autek Research RX Vector Analyst VA1. Although both do roughly the same thing, they have a few different features and I will touch on a few of those.
Besides digital readout, the MFJ 259B also has an analog SWR meter which greatly helps in finding resonant spots on an antenna using a frequency sweep. You can watch if you are getting close or drifting away much easier than with digital readout.
You can measure capacitance and inductance with both but the MFJ259B does not give you the sign of the reactance. That is, negative for capacitance and positive for inductance. You can determine the sign merely by doing a small shift in frequency and seeing which direction the reactance swings (lower for capacitance if frequency shifts upward for example).
The Autek VA1 though includes the arithmetic sign of reactance so it is just slightly easier if you want a capacitance versus inductance readout.
So, whether you are building antennas for resonance or cutting transmission line to be a transformer, the use of either of these two analyzers makes it gut simple.
Postscript: I forgot to mention whether an SWR meter is any good? Well, it can tell you when you are really bad but it does not really help enough in my opinion. After all, a 50-ohm resistant dummy load gives an SWR of 1:1 but that does not say it is a good antenna. Also, by knowing the R and X values of the reactance, you are able to determine where problems might lie with an antenna. Remember that resonance is where it is purely resistive so $X=0$ or close to zero.
$\begingroup$ Minimum SWR is not equivalent to resonance. Resonance is where voltage is in phase with current, which may or may not be at minimum SWR. So, sweeping with an SWR meter does not find resonance. $\endgroup$ Jan 6, 2015 at 21:54
$\begingroup$ @PhilFrost -- Voltage is in phase with current when reactance is zero. Which is what I said. I also said that a purely resistive (reactance zero) impedance does not necessarily mean that you have a good antenna as that resistance can be totally loss resistance (i.e. zero radiation resistance) such as the example I gave of a dummy load. I don't think I ever said resonance was at minimum SWR. I did say that watching the SWR on the analog meter was helpful in finding resonant spots and it is. The readout also tells you R and X. You know when it is resonant! $\endgroup$– K7PEHJan 6, 2015 at 22:38
1$\begingroup$ The particular confusing statement is "the MFJ 259B also has an analog SWR meter which greatly helps in finding resonant spots on an antenna using a frequency sweep." I realise that doesn't directly say minimum SWR is resonance, but it pretty strongly implies it. Maybe you could edit to make it more clear, incorporating what you have said in your comment. $\endgroup$ Jan 6, 2015 at 23:50
I've been a ham since 1963 and have built all my own antennas. Yagi beams—wire beams—loops—verts, etc. Back in those days all you had was a SWR bridge and maybe a grid dip meter. The most important thing, I think, is a booklet or some kind of articles on antenna design. They will give you accurate lengths and feed line info that should get you up and running in no time. There's plenty of info on the internet, although my personal favorite source is Bill Orr, W6SAI; his antenna books and very entertaining antenna articles in CQ magazine circa 1970's are great.
My first antenna was made of electric fence wire and a second hand hunk of RG-58 in an inverted vee configuration. It is supremely simple to build. One support pole, one hunk of 50 ohm coax, balun or not, 120 degree spread between the legs, gives you 50 ohms. Make it about three feet longer than formula and trim it to resonance using an el cheapo used (\$10) SWR bridge.
A used grid dip meter (\$30.00) will let you get the resonant frequency—not necessary here—and a used (\$30.00) noise bridge will let you deal with impedance issues, but is not necessary with simple antennas like this. Simple antennas are easy to build and work well. Height is a big plus. Just get in there and klutz around until it works. You can build anything easily with the three pieces of test gear mentioned here and they will do a lot more things than help set up antennas. Simplicity is bliss.
After looking at several wire antennas decided to build my own and save money plus having the satisfaction of knowing I built a successful antenna. Wire, insulators, solder and rope plus sling shot to elevate antenna at a reasonable height. Had plenty of room for dipoles or loops. 468 divided by frequency will give you a usable wire antenna. Different gauges of wire will give different results. For those with limited space coils will shrink physical size to fit your space with some trade-off. Rotatable dipoles for 10, 20 meters are easy to build with nesting aluminum tubing. Antenna analyzers help for those who want a more accurate deigns but not necessary. My 560 foot loop for 160 out performed beams and at time 100 watts was enough to beat amps.
1$\begingroup$ Hello Fred and welcome to ham.stackexchange.com! Your post is interesting and relevant, but doesn't completely answer the question. How does someone use 468/f? Would it work for all kinds of antennas? $\endgroup$ Aug 31, 2020 at 15:41